## Seebeck Voltages

I have an issue with converting these to temperatures. I have the equation, voltage is in microvolts, but when I convert, say boiling water at 3600 microvolts to celsius, it gives me something like 90 C. It ought to be close to 95 C, I am in Colorado.

I did it today in excel, and the matlab equation I wrote for it last week gave me the same answer. According to the lab handouts, the error should only be +/- 0.05 C.

Thanks for any help!
 PhysOrg.com science news on PhysOrg.com >> City-life changes blackbird personalities, study shows>> Origins of 'The Hoff' crab revealed (w/ Video)>> Older males make better fathers: Mature male beetles work harder, care less about female infidelity
 Recognitions: Science Advisor Let's see what you are doing. Are you actually using a thermocouple? What metals are you using? You could possibly be using the wrong coefficients...
 Yes, we are using thermocouples made of Ni-Ch and Ni-Al Type K thermocouples (given by profs) Here is the equation T = 0.0 + 2.508355e-2*v + 7.860106e–8*v^2 + –2.503131e–10*v^3 + 8.315270e–14*v^4 + –1.228034e–17*v^5 + 9.804036e–22*v^6 + –4.413030e–26*v^7 + 1.057734e–30*v^8 + –1.052755e–35*v^9 Again v is in microvolts, and T is in celsius. Because I am just a few degrees off, I am wondering if the polynomial is right, ie do we have the right thermocouple type.

## Seebeck Voltages

anyone else?

I have checked this in matlab, and with other people, getting the same answer.

but the LabVIEW VI software in the lab gives us the right answer
 Recognitions: Science Advisor Looking at my ASTM tc reference, the K type polynomial is different than what you have. Give me a bit and I'll post it.
 OK, thanks. I pulled mine from National Instruments documentation on thermocouples, but I will be interested to see yours.
 Recognitions: Science Advisor Without running some numbers I can't say if this will fix your offset, but I did forget that there are two exponential corrections for the first two terms. Take a look at the attached page. Also note the reference temperature of 0° Ref. ASTM Manual on the use of thermocouples in temperature measurement, 4th edition.
 Thanks for that, but the problem with that is that converts from temp to volts, i need to go the other way.
 Recognitions: Science Advisor All right. Let me see what else I can dig up.
 Recognitions: Science Advisor OK. Try this one. Hopefully this one will help. This is from NIST. It is a typical K type TC table, however at the very end there is a section for the inverse function of temp and emf. Give that a look see and maybe that will hlp you out.

 Similar discussions for: Seebeck Voltages Thread Forum Replies Classical Physics 2 General Physics 0 Classical Physics 1 General Physics 0 General Physics 2