# How to measure parasitic capacitance of inductor

• learner1
In summary: If you are measuring capacitance in a circuit, you will need to provide a means of tuning the inductor.
learner1
How can I get the parasitic capacitance of a inductor? I have a scope, current probe, function generator.

Thank you.

What is the value of incuctance?

Pretend the inductor is a capacitor, and measure its capacitance at high frequencies.

You can do it manually by simply plotting a frequency/Impedance graph, and curve-fitting some sample values of capacitance based upon the 1/2piefc algebraic equation.

You can connect your inductor to the signal generator through a resistor and measure the voltage across the inductor with your scope.
Varying the frequency of your sinusoid will change the amplitude of the signal. The amplitude will be minimal at the resonant frequency $$\omega_0$$.
Since $$\omega_0 = \frac{1}{\sqrt{LC}}$$ you can calculate the capacitance.

learner1 said:
How can I get the parasitic capacitance of a inductor? I have a scope, current probe, function generator.

Thank you.
The easiest way will be from its datasheet. There should be a DCR (series resistance) and parallel capacitance listed. Depending on the L and C values, you may have a hard time measuring the resonance with your signal generator. Like if the sig gen only goes to 10MHz, and your inductor is 10uH with 10pF of capacitance, your resonant freq will be around 16MHz. Also keep in mind that your oscilloscope probe has capacitance (probably in the 10pF range), so that will alter your reading and needs to be accounted for.

If you need higher frequencies for your tests, you can try adding a series diode and driving the inductor with a square wave. You would make a low-inductance measuring fixture similar to a Z-lead for your scope probe, and measure the ringout frequency of the inductor and parasitic capacitance (and then subtract the scope probe capacitance and any other fixture capacitance).

Let us know what you come up with, and maybe post a link to the inductor's datasheet so we can check it out. -Mike-

Thanks for all the suggestions. I tried SRF one, and here is the result.
I know the inductance is about 1.4mH. The inductor was directly connected to the signal generator. The voltage and current across the inductor are measured using scope. I increased the frequency of sinusoid until phase between V and I changed from 90 to –90 (impedance from inductive to capacitive). Then I use this frequency (420kHz) to calculate C based on . C is about 100pH.
Is this value reasonable?
I want to try something else to verify it.

It is totally unreasonable because capacitance is not measured in pH. :rofl: I suspect you mean pF. Anyway, seriously I think it is probably correct. Your approach is the way I would do it.

Yes, pF.

I am thinking about another way to measure it. If the output impedance of the function generator is 50ohms, can I use Xc=50=1/2pifC and find this f? Am I looking for -45 phase shift? Thank you.

I would use a series resistor and note the frequency at resonance and several other places. Make sure your calculations all come up with about the same capacitance value.

keep in mind the capacitance effects you are measuring should be small, but drastically effected by the presence of nearby objects. So do you want to measure it in isolation, or in the actual circuit you intend to use it? This is not mere philosophy. In RF circuits it is critical, and usually some method of 'tuning' the inductor must be provided.

## 1. What is parasitic capacitance?

Parasitic capacitance refers to the unintentional or unwanted capacitance that exists between different components or elements in a circuit.

## 2. Why is it important to measure parasitic capacitance of inductors?

Measuring parasitic capacitance of inductors is important because it can significantly affect the performance and stability of a circuit. It can also cause unexpected behavior and interference in the circuit.

## 3. What equipment is needed to measure parasitic capacitance of inductors?

The most common equipment used to measure parasitic capacitance of inductors is a capacitance meter or an LCR meter. These devices can accurately measure the capacitance of the inductor and other components in a circuit.

## 4. How do you measure parasitic capacitance of inductors?

To measure parasitic capacitance of inductors, the inductor is connected to a capacitance meter and the capacitance value is recorded. The inductor should be isolated from other components and any stray capacitance should be accounted for in the measurement.

## 5. How can parasitic capacitance of inductors be reduced?

To reduce parasitic capacitance of inductors, the distance between components should be increased and the use of low parasitic materials, such as air-core or thin film inductors, can also help. Proper grounding and shielding techniques can also minimize parasitic capacitance in a circuit.

• Electrical Engineering
Replies
36
Views
3K
• Electrical Engineering
Replies
17
Views
2K
• Electrical Engineering
Replies
14
Views
2K
• Electrical Engineering
Replies
17
Views
1K
• Electrical Engineering
Replies
32
Views
2K
• Electrical Engineering
Replies
28
Views
2K
• Electrical Engineering
Replies
14
Views
3K
• Electrical Engineering
Replies
5
Views
840
• Electrical Engineering
Replies
10
Views
459
• Electrical Engineering
Replies
13
Views
2K