# Seebeck Voltages

I have an issue with converting these to temperatures. I have the equation, voltage is in microvolts, but when I convert, say boiling water at 3600 microvolts to celsius, it gives me something like 90 C. It ought to be close to 95 C, I am in Colorado.

I did it today in excel, and the matlab equation I wrote for it last week gave me the same answer. According to the lab handouts, the error should only be +/- 0.05 C.

Thanks for any help!

Related Introductory Physics Homework Help News on Phys.org
FredGarvin
Let's see what you are doing. Are you actually using a thermocouple? What metals are you using? You could possibly be using the wrong coefficients...

Yes, we are using thermocouples made of Ni-Ch and Ni-Al

Type K thermocouples (given by profs)

Here is the equation

T = 0.0 + 2.508355e-2*v + 7.860106e–8*v^2 + –2.503131e–10*v^3 + 8.315270e–14*v^4 + –1.228034e–17*v^5 + 9.804036e–22*v^6 + –4.413030e–26*v^7 + 1.057734e–30*v^8 + –1.052755e–35*v^9

Again v is in microvolts, and T is in celsius.

Because I am just a few degrees off, I am wondering if the polynomial is right, ie do we have the right thermocouple type.

anyone else?

I have checked this in matlab, and with other people, getting the same answer.

but the LabVIEW VI software in the lab gives us the right answer

FredGarvin
Looking at my ASTM tc reference, the K type polynomial is different than what you have. Give me a bit and I'll post it.

OK, thanks.

I pulled mine from National Instruments documentation on thermocouples, but I will be interested to see yours.

FredGarvin
Without running some numbers I can't say if this will fix your offset, but I did forget that there are two exponential corrections for the first two terms. Take a look at the attached page.

Also note the reference temperature of 0°

Ref. ASTM Manual on the use of thermocouples in temperature measurement, 4th edition.

Last edited:
Thanks for that, but the problem with that is that converts from temp to volts, i need to go the other way.

FredGarvin