Register to reply 
Vibration profile calculation for shaker table 
Share this thread: 
#1
Feb413, 12:44 PM

P: 3

Hi guys,
I hope we got some expert out here for vibration profile problems. I have a device that need to be on a shaker table but I need to "find" its profile to see if the shaker table can actually run this experiment. The Shaker table is UnholtzDickie system SA30560B and I have data of this table's profile in Acceleration (g PK) vs Frequency (Hz). My device with fixture weights 23.5lbs, and I need to run it between frequency of 10 to 500 Hz with 30.59g force, this is a long life test. Can someone help me on finding the X, X2, X3? Also that ASD stands for what? I can't figure out what that stands for, but the table's profile has in Acceleration of gforce Peak vs Frequency instead of ASD vs. Frequency. 


#2
Feb413, 04:16 PM

Engineering
Sci Advisor
HW Helper
Thanks
P: 6,948

ASD = acceleration spectral density (similar to power spectral density).
The values of x, x2, x3 are all defined by the graph and the table of values underneath. If you don't understand how to interpret the data, I think you need to review the basics of random vibrration theory and testing. There is no way you will get 30g at 500Hz from that ASD specification. You also need to check the force output is big enough to shake the mass (plus the mass of the table, etc) at 30g. This http://www.acutronic.com/uploads/med...tEquipment.pdf suggests the continuous power rating of the SA30 amp isn't big enough. 


#3
Feb513, 06:48 AM

P: 3

Again don't worry about the fixture weight for now, can you tell me how you can calculate the PSD x, x2 and x3? I need those 3 values to compare it to the shaker table's profile and see at which point they overlap. Thanks. 


#4
Feb613, 07:34 AM

P: 3

Vibration profile calculation for shaker table
Bump, anyone able to help me?



#5
Mar413, 10:19 AM

P: 7

I found the answer, in case anyone else like to know how to solve it.
I was giving an ASD value which represented value in mid (flat) line of profile, I used db/Oct and Hz, along with ASD flat line value with this equation: P1=P2*(f1/f2)^((db/oct)/3) P1 is value of ASD I'm looking for at f1=10 Hz While P2 is the given ASD value of flat line of profile. Also, if the value of db/Oct is negative then you will use this: P1=P2/((f1/f2)^((db/oct)/3)) 


Register to reply 
Related Discussions  
Mechanical shaker for vibration testing  General Engineering  1  
Vibration shaker table  General Physics  0  
Vibration shaker: electr. output power of piezodevice exceeds mechanical input power  Advanced Physics Homework  0  
Vibration table experiment  Engineering, Comp Sci, & Technology Homework  0  
Vibration Free Table for Holograms  General Physics  16 