- #1

- 83

- 13

I measuring current with a Rogowski Coil and I want to perform numerical integration on the output.

Since the output is fluctuating around ground level, the question is: Should I use the absolute values of the output or not?

Thank you.

- A
- Thread starter Abimbola1987
- Start date

- #1

- 83

- 13

I measuring current with a Rogowski Coil and I want to perform numerical integration on the output.

Since the output is fluctuating around ground level, the question is: Should I use the absolute values of the output or not?

Thank you.

- #2

DaveE

Gold Member

- 993

- 739

- #3

CWatters

Science Advisor

Homework Helper

Gold Member

- 10,529

- 2,297

More specifically for power you want to integrate the product of the instantaneous Voltage and instantaneous Current.

If the voltage is constant you can take that outside the integral. Eg integrate the absolute current then multiply by the voltage.

- #4

- 83

- 13

Thank you and duly noted, however I havn't been able to measure any DC current using the shunt resistor method. Also it is my understanding that after integrating the output voltage, it has to be scaled by a factor of 1/M, where M is the mutual inductance between the coil and wire being measured.The coil will not respond to the constant (DC) part of the current being measured. So the coil output will be an AC voltage and the current being measured will be the integral of this voltage with the DC part removed, also AC. You may want to "add in" the DC via analysis of the circuit or a different DC current measurement.

What do you mean by "true wave shape"? the instantaneous current?In any case if you perform an absolute value you will distort the true wave shape (rectification). I suppose you may want to do that, but I doubt it.

And as CWatters noted in the post above, I will be using the value to calculate power.

- #5

- 83

- 13

Thank you and duly noted. So when would I

More specifically for power you want to integrate the product of the instantaneous Voltage and instantaneous Current.

If the voltage is constant you can take that outside the integral. Eg integrate the absolute current then multiply by the voltage.

- #6

CWatters

Science Advisor

Homework Helper

Gold Member

- 10,529

- 2,297

If you are trying to measure the power dissipated in a diode that is part of a rectifier then it would only be (for example) the positive part of the cycle you use because the negative half flows through another diode.

- #7

Tom.G

Science Advisor

- 3,536

- 2,271

Designing with them is not all that simple. See https://www.physicsforums.com/threads/adder-lp-filter.943146/

The last post in that thread lists some sources for commercially available instruments.

Cheers,

Tom

- Replies
- 11

- Views
- 5K

- Last Post

- Replies
- 11

- Views
- 3K

- Last Post

- Replies
- 12

- Views
- 1K

- Last Post

- Replies
- 2

- Views
- 14K

- Last Post

- Replies
- 9

- Views
- 2K

- Replies
- 3

- Views
- 656

- Last Post

- Replies
- 8

- Views
- 1K

- Replies
- 1

- Views
- 2K

- Replies
- 4

- Views
- 1K

- Replies
- 22

- Views
- 2K